James A. Zagzebski, Ph.D., Professor Emeritus for the University of Wisconsin – Madison Dept. of Medical Physics, demonstrates how to use the Sono403 and SonoTE Phantoms in support of ACR QA and accreditation requirements. Watch his series of how-to videos below.
Shadows or artifacts in the image indicate a problem with one or more transducer elements, while the following image characteristics indicate the transducer is functioning properly:
Personalize the Sono TE Phantom with your brand. You may select from a range of colors and add your logo1.
HE Gel™: Gammex’s multi-frequency tissue mimicking material
|Dimensions (L/W/H):||11.5 x 5.7 x 7.5 cm (4.63 x 2.25 x 3 in)|
|Weight:||580 +/- 5 g (1 lb. 4 oz.)|
Designed for use with linear, convex and inter-cavity transducers. Not for use with phased array sector scan transducers.
1 Personalized for large orders.